M6013 V2 Autorange Capacitor Capacitance Tester Meter Cap 0.01pF to 470mF.It is an Auto-ranging capacitance meter.It has fast reading speed and high accuracy up to 1%.Stable on reading, (with auto running averaging function, can read pF more stable and accuracy)
M6013 V2 Autorange Capacitor Capacitance Tester Meter Cap 0.01pF to 470mF
TOP 4 Reasons to get M6013:
1. JINGYAN M6013 V2 Auto-ranging capacitance meter.
2. M6013 have fast reading speed and high accuracy up to 1%.
3. Reading a 2200uF only take ~0.2s
4. Using 128X64 dot matrix LCD, can display 5 digit reading, Large range of measuring 0.01pF to 470mF (470000uF) capacitance,
which enough for DIY user to professional engineer usage.
Note:Please Discharge The Capacitor Before Testing
Compare M6013 old V1 and new V2 Improvement:
1. Larger measuring range, V2 can read up to 470mF
2. New large dot matrix LCD
3. New housing with stand
4. More stable on reading, (with auto running averaging function, can read pF more stable and accuracy)
5. Can use 2X AA battery rather than 9V
6. Can use mirco usb as external power.
Accuracy (After Zero, tested with 1nF, 1uF, 1000uF)
Refresh Time (Manual Mode)
*Auto mode take 0 to 2s more time depend on value
Larger capacitance take more measuring time
0.01pF to 47.000nF
~0.2s to 1s
47.00nF to 47.000uF
~0.2s to 2s
47.00uF to 470.00mF
1 to 3%+1Digit (Larger capacitance will have larger error)
~0.2s to 30s
Up to 1% (detail on above table)
~1.25V (open voltage)
2X AA 1.5V battery
5V micro USB
Battery Life time
Note :Accuracy maybe affected by the test lead’s length and distance of test leads. Especially testing pF small capacitance, shortest test
lead is recommended, and be careful the surrounding EMI or RF noise may affect the pF reading.
1) Poqwer On
Press and hold the ON/ OFF circle orange button for 1 to 2 second to power on.
Press and release the ON/ OFF button to shut down meter.
2) Auto/ Manual mode:
i) Auto Range:
Press and release the RANGE button and at the first line of LCD will show “AUTO:"
At auto mode meter will automatically select the best range to detect.
ii) Manual Range:
Scroll the manual range from 47nF, 47uF and 470mF range by press and release the RANGE button
In LCD will show MANUAL at first LCD line and show at second line: 0-47NF, 47NF TO 47UF and 47UF TO 470MF.
3) 1 Key Zero:
OPEN circuit the test leads’ terminal.
Press and release the “ZERO" button, LCD shows “ZERO" and wait the zero disappear.
If you are using the meter array socket, you need to OPEN circuit to set zero too.
You can use the 8 pin socket to measure the capacitor
5) Running Average:
It will automatically start running average if the capacitance reading become stable and user can get more accuracy reading.
During running average it will show “AVG” at right bottom LCD or it will show raw data icon “RAW”
As a result, if you want to get higher accuracy reading, you can take the reading during AVG display.
This function will automatically on, once the raw data is noise.
LCD backlight will on during power on
7) Auto Sleep:
Around 8 to 10 hours for not testing, it will shut down automatically to save power.
8) Overflow or Ol:
Display overflow when the value is out of range, you can check that you are zero correctly.
9) Plstic Stand:
If you don’t use the stand, press and push the lock position:
M6013 Package Including:
1unit x M6013
1pair x test probe
If you have any questions, please feel free to contact us by following way:
Whatsapp: 0086-18046220975Shipment Warm Tips:
We only ship to the address on Paypal. If you require a specific address to ship the order to, please make sure that you enter this in the Paypal confirmation before confirming payment. We will not ship to any other address under any circumstances. If you have asked for a change in address please add your shipping address on your paypal account,please leave a message on paypal when you check out, or send us the screen shot of new shipping address under paypal account
to sales@iDiyTool.com or whatsapp:+86 18046220975.